Most likely they were testing a live circuit. Continuity testers are designed to test UNENERGIZED circuits. The bulbs in these testers are only rated for around 3 volts. If you hook this up to a live circuit in your house, you just put 120 volts through the bulb and that is why it blew out.
No, we do not offer leads for these. The way these are manufactured, you would have to disassemble the entire tester and re-solder new leads on. These testers are not designed to be taken apart.
These testers were designed for quick basic testing to determine if there is voltage at an outlet or on a line. They are not designed to pin-point which line is the hot and which is neutral. For that you would want to use the VD6505 because they have an adjustment for the sensitivity to dial it down and give more accurate readings.
No, we do not have replacement lenses or decals for the outlet testers. Any of the testers that have lenses that would fall out have been obsolete for about 2 years.
We do not have any type of covers or protective cases for the meters. Some units will come with a case, but we do not sell them individually.
Yes, for the GB and Sperry meters RTL-107 is a slide on alligator clip that slides on to the metal tip of the test leads. For a customer looking for Sperry leads with alligator clips, TL42 has removable alligator clips. The alligator clips that come with these leads will not work with any other set of leads because instead of just sliding on the test leads, these ones will screw onto the lead.
Any multimeter with the ability to test resistance or ohms can measure for continuity. Some meters have an audible buzzer to let you know when you have continuity, while others don’t. You need to keep an eye on the display to watch the reading the tester is giving. Most analog testers you have to watch the screen to see if the needle goes all the way to the right. With the digital meters, some have an audible symbol to let you know if the meter has an audible buzzer or not.
This is the audible continuity buzzer. In most cases, if the reading is under 50 ohms of resistance, the meter will buzz. With other testers, you would just select any of the resistance ranges (Ω) and watch the reading you get on the screen. It should be as close to 0.00 as possible to have the best continuity. Most times you will get something between 0.00 and 0.3 on the display because the meter is picking up resistance in the circuit being tested and in the test leads.
WARNING!!! ALWAYS TEST CONTINUITY ON A CIRCUIT THAT IS SHUT OFF. A live circuit will damage the tester and potentially injure the user.
If you find yourself having trouble keeping the CS61200 turned off while in your tool bag. Than this simple fix may be the trick you’ve been looking for. In the below picture you’ll see the CS61200 and you’ll notice something a little different. One of our customers attached two o-rings to the front of the unit guarding the on/off button. This keeps the unit from turning itself on while in your tool bag.
On many of our Sperry meters there is a hold button that will freeze the screen, holding the information of the last reading you took. When this option is in use, there is a small letter “H” displayed on the screen. This allows you to write down the reading without having to hold the test leads in place.
If you’re having trouble taking readings and the readings on the screen do not change, look for the H. This may be your problem. In other cases it’s due to a low battery, which causes the meter to give faulty readings.
Each meter has a specific range for accuracy; this is always listed in the unit’s manual. DC voltage will always have a better accuracy rating because its electrical wave form does not fluctuate like AC voltage.